This instrument is designed and manufactured as per standards GB/T 261-2008 Determination of Flash Point - Pensky-Martens closed cup method and ASTM D93 Standard Test Methods for Flash Point by Pensky-Martens Closed Cup Tester.
Technical Data Sheet
I. Main technical features
1. Single chip microcomputer control technique.LCD screen displays.English man-machine interaction interface.The heating power is continuously step-less adjustable.The rate of heating up is exact.Easy setting for parameters.
2. The LCD screen has prompting menu,prompting input operation interface.It shows the setted parameters,real-time temperatures and other test parameters.Press the “record”button when the flash point appears.The screen will show and reserve the flash point.
3. The structure is newly design and small.Stainless steel table-board looks clean and beautiful.It can be used as long as equipped with gas or other civilian fuels.It is easy to use and test results exact.
|1||Power supply:||AC 220V±10%, 50 Hz;|
|2||Heating device|| |
|2.1||Furnace power is; itsis||600 W， made of SiC materials|
|2.2||Heating power adjustable range||0 W～600W|
|3||Sample temperature rising rate||Procedure A ：(5～8 )℃/min; Procedure B：(1～1.5)℃/min；Automatically control|
|4||Electric stirring device|| |
|4.1||Stirring motor||BYGH101stepping motor|
|4.2||Driving mode||Flexible axle|
|4.3||Stirring blade specification||8 mm×40 mm|
|5||Stirring speed||As per the requirement of GB/T261-2008|
|5.1||Procedure A;||90 RPM～120 RPM|
|5.2||Procedure B;||250±10 RPM|
|6||Standard oil cup||As per the requirement of GB/T261-2008|
|6.1||Inner diameter||50.7 mm～50.8 mm|
|6.2||Depth||55.7 mm～56.0 mm|
|6.3||Depth of marked line in the oil cup||33.9 mm～34.3 mm|
|6.4||Oil sample cubage||About 70 ml|
|7||Flame application device|| |
|7.1||Gas source||Gas (or other civilian fuel,the same below)|
|7.2||Aperture of flame application device||0.8 mm|
|11||Total power consumption||Not more than 650 W|